Payment & Shipping Terms:
|Delivery:||Within 48 Hours||Test Method:||Sandwich|
elisa assay kit,
Human Customized RBP-4 ELISA Test Kit Retinol Binding Protein 4 ELISA Assay Kit With Strong Sensitivity
Serum Allow serum to clot for 10-20 minutes at room temperature. Centrifuge at 2000-3000 RPM for 20 minutes.
Plasma Collect plasma using EDTA or heparin as an anticoagulant. Centrifuge samples for 15 minutes at 2000-3000 RPM at 2 - 8°C within 30 minutes of collection.
Urine Collect by sterile tube. Centrifuge at 2000-3000 RPM for approximately 20 minutes. When collecting pleuroperitoneal fluid and cerebrospinal fluid, please follow the procedures above-mentioned.
Cell Culture Supernatant Collect by sterile tubes when examining secrete components. Centrifuge at 2000-3000 RPM for approximately 20 minutes. Collect the supernatants carefully. When examining the components within the cell, use PBS (pH 7.2-7.4) to dilute cell suspension to the cell concentration of approximately 1 million/ml. Damage cells through repeated freeze-thaw cycles to let out the inside components. Centrifuge at 2000-3000 RPM for approximately 20 minutes.
Tissue and other body fluids Rinse tissues in PBS (pH 7.4) to remove excess blood thoroughly and weigh before homogenization. Mince tissues and homogenize them in PBS (pH7.4) with a glass homogenizer on ice. Thaw at 2-8°C or freeze at -20°C. Centrifuge at 2000-3000 RPM for approximately 20 minutes.
*Sample can't be diluted with this kit. Owing to the the material we use to prepare the kit, the sample matrix interference may falsely depress the specificity and accuracy of the assay.
Standard Curve Range: 1ng/ml - 400ng/ml
Size: 96 wells
*This product is for research use only, not for use in diagnosis procedures. It’s highly recommend to read this instruction entirely before use.
|Standard Solution (480ng/ml)||0.5ml x1|
|Pre-coated ELISA Plate||12 * 8 well strips x1|
|Standard Diluent||3ml x1|
|Stop Solution||6ml x1|
|Substrate Solution A||6ml x1|
|Substrate Solution B||6ml x1|
|Wash Buffer Concentrate (25x)||20ml x1|
|Biotinylated human RBP-4 Antibody||1ml x1|
|Plate Sealer||2 pics|
|Zipper bag||1 pic|
All reagents should be brought to room temperature before use.
Standard Reconstitute the 120μl of the standard (480ng/ml) with 120μl of standard diluent to generate a 240ng/ml standard stock solution. Allow the standard to sit for 15 mins with gentle agitation prior to making dilutions. Prepare duplicate standard points by serially diluting the standard stock solution (240ng/ml) 1:2 with standard diluent to produce 120ng/ml, 60ng/ml, 30ng/ml and 15ng/ml solutions. Standard diluent serves as the zero standard(0 ng/ml). Any remaining solution should be frozen at -20°C and used within one month. Dilution of standard solutions suggested are as follows:
|240ng/ml||Standard No.5||120μl Original Standard + 120μl Standard Diluent|
|120ng/ml||Standard No.4||120μl Standard No.5 + 120μl Standard Diluent|
|60ng/ml||Standard No.3||120μl Standard No.4 + 120μl Standard Diluent|
|30ng/ml||Standard No.2||120μl Standard No.3 + 120μl Standard Diluent|
|15ng/ml||Standard No.1||120μl Standard No.2 + 120μl Standard Diluent|
|Standard Concentration||Standard No.5||Standard No.4||Standard No.3||Standard No.2||Standard No.1|
Wash Buffer Dilute 20ml of Wash Buffer Concentrate 25x into deionized or distilled water to yield 500 ml of 1x Wash Buffer. If crystals have formed in the concentrate, mix gently until the crystals have completely dissolved.
1. Prepare all reagents, samples and standards.
2. Add sample and ELISA reagent into each well. Incubate for 1 hour at 37°C.
3. Wash the plate 5 times.
4. Add substrate solution A and B. Incubate for 10 minutes at 37°C.
5. Add stop solution and color develops.
6. Read the OD value within 10 minutes.
Calculation of Result
Construct a standard curve by plotting the average OD for each standard on the vertical (Y) axis against the concentration on the horizontal (X) axis and draw a best fit curve through the points on the graph. These calculations can be best performed with computer-based curve-fitting software and the best fit line can be determined by regression analysis.
Contact Person: Lee