Payment & Shipping Terms:
|Assay Time:||2 Hours||Target Protein:||Anti-Mullerian Hormone|
|Standard Curve Range:||0.1ng/ml - 40ng/ml||Sensitivity:||0.05ng/ml|
96 wells Mouse Anti-Mullerian Hormone AMH ELISA Kit High Sensitivity
Standard Curve Range: 0.1ng/ml - 40ng/ml
Size: 96 wells
Storage: Store the reagents at 2-8°C. For over 6-month storage refer to the expiration date keep it at -20°C. Avoid repeated thaw cycles. If individual reagents are opened it is recommended that the kit be used within 1 month.
* This product is for research use only, not for use in diagnosis procedures. It’s highly recommend to read this instruction entirely before use.
This sandwich kit is for the accurate quantitative detection of Mouse Anti-Mullerian Hormone (also known as AMH) in serum, plasma, cell culture supernates, cell lysates, tissue homogenates.
Intra-Assay Precision (Precision within an assay) Three samples of known concentration were tested on one plate to assess intra-assay precision.
Inter-Assay Precision (Precision between assays) Three samples of known concentration were tested in separate assays to assess inter-assay precision.
CV(%) = SD/mean x 100
All reagents should be brought to room temperature before use.
Standard Reconstitute the 120μl of the standard (48ng/ml) with 120μl of standard diluent to generate a 24ng/ml standard stock solution. Allow the standard to sit for 15 mins with gentle agitation prior to making dilutions. Prepare duplicate standard points by serially diluting the standard stock solution (24ng/ml) 1:2 with standard diluent to produce 12ng/ml, 6ng/ml, 3ng/ml and 1.5ng/ml solutions. Standard diluent serves as the zero standard(0 ng/ml). Any remaining solution should be frozen at -20°C and used within one month. Dilution of standard solutions suggested are as follows:
|24ng/ml||Standard No.5||120μl Original Standard + 120μl Standard Diluent|
|12ng/ml||Standard No.4||120μl Standard No.5 + 120μl Standard Diluent|
|6ng/ml||Standard No.3||120μl Standard No.4 + 120μl Standard Diluent|
|3ng/ml||Standard No.2||120μl Standard No.3 + 120μl Standard Diluent|
|1.5ng/ml||Standard No.1||120μl Standard No.2 + 120μl Standard Diluent|
|Standard Concentration||Standard No.5||Standard No.4||Standard No.3||Standard No.2||Standard No.1|
Wash Buffer Dilute 20ml of Wash Buffer Concentrate 30x into deionized or distilled water to yield 500 ml of 1x Wash Buffer. If crystals have formed in the concentrate, mix gently until the crystals have completely dissolved.
Calculation of Result
Construct a standard curve by plotting the average OD for each standard on the vertical (Y) axis against the concentration on the horizontal (X) axis and draw a best fit curve through the points on the graph. These calculations can be best performed with computer-based curve-fitting software and the best fit line can be determined by regression analysis.
Durlinger A.L., Visser J.A., Themmen A.P.
Contact Person: Lee